8″ H x 8″ W x 6″ D, excluding pole clamp
9.5 lbs. with battery
Delivery Rate Range
Lines A & B:
0.1-99.9 ml/hr in 0.1 ml increments100-999 ml/hr in 1 ml increments
0.5 ml/hr minimum for each line
500 ml/hr cumulative (A+B) maximum
1.0 ml/hr or the last primary delivery rate, whichever is less
0.1-99.9 ml in 0.1 ml increments100-9999 ml in 1 ml increments
Electrical Power Requirements
120V, 50-60 Hz, 35VA
F1, F2, 250V, 0.5A
Hospital Grade AC Cord 10 ft long
One sealed lead acid, rechargeable 6V battery, internal
With a new fully charged battery, the infuser shall operate for a minimum of 6 hours at 125 ml/hr or less, or deliver 500 ml at 126 ml/hr or greater on one line.
The battery charges whenever the pump is connected to AC power. The recharge time is approximately 6 hours with the device operating at 125 ml/hr on one line.