Linear Peristaltic Volumetric Infusion Pump
Keep Vein Open (KVO) Rate
5 mL/hr or programmed rate, whichever is less
Available through the COMMUNICATIONS PORT
12 Volt, 2.0 Ah sealed lead-acid
Approximately 6 hours with pump running at a rate from 1 to 1400 mL/hr
8 hours for more than 80% recharge
AC Power Requirements
110/120V, 60 Hz
2.9 m (9 ft) long, with hospital grade plug
0.5 A, 250V, SB, 5.2 mm (13/64 in) x 20 mm (25/32 in)
Less than 50 microamps (using UL-544 specified test methods)
Approximately 5.3 kg (11.6 lbs)
20 cm W x 13 cm D x 29 cm H (7.9″ W x 5.1″ D x 11.4″ H)
Flow Rate Range
Primary program: 1.0-99.9 mL/hr in 0.1 mL increments or 1-1999 mL/hr in 1 mL increments. Upper limit can be reduced by authorized service personnel.
Secondary program: 1.0-99.9 mL/hr in 0.1 mL increments or 1-999 mL/hr in 1 mL increments
1.0-99.9 mL in 0.1 mL increments or 1-9999 mL in 1 mL increments for both primary and secondary. Upper limit can be reduced by authorized service personnel.
Ramp PDP Parameter Ranges
Total VTBI: 100-9,999 in 1 mL increments. Upper limit can be reduced by authorized service personnel.
Delay Time: 00:00 – 08:00 hours in minute increments.
Run Time: 01:00 – 16:00 hours in minute increments.
Ramp Up: 00:00 to 45% of the Run Time, in minute increments.
Ramp Down: 00-00 to the Run Time minus Ramp Up, in minute Increments.
Factory set to NORM, which causes the pump to alarm on air bubbles approximately 75 uL or larger. The MIN setting causes the pump to alarm on air bubbles approximately 50 ul or larger.